As a key system in hospital decoration, purification system is the key element to control and protect the infection risk of patients and medical staff. But in the long-term use, there will inevitably be some problems. So today, let's talk about the possible problems and solutions of the operating room decoration and purification system?
As we all know, the purification system used in hospital decoration is formed by the combination of clean technology and clean air conditioning. However, if the operating room purification system is used for a long time, there may be blackening, and there are many reasons for the oxidation blackening.
The common copper pipe welding is gas welding and fiber welding. However, the phenomenon of blackening often appears after welding. In fact, the physical properties of the pipeline have changed and oxidation has occurred, which has an irreparable impact on the life of the pipeline and the quality of oxygen. Therefore, the pipeline should be protected when welding, and most of the welding will use gas to protect the welding process from oxidation.
When the pipeline is welded, nitrogen is slowly blown into the other end of the pipe to protect the pipeline. With the protection of nitrogen, the pipeline after welding is as bright as before, without any blackening and oxidation. It is a good and effective method to solve the oxidation problem of pipeline in the purification system of operating room.
Today, we will introduce the problems that may arise in the decoration and purification system of the operating room of the hospital. If you want to know more about the decoration of the hospital, the decoration of the operating room and other decoration, you can pay attention to our website: http://www.sdyhrs.com