Operation shadowless lamp is indispensable for laminar flow operating room decoration! Now let's Take Shandong jingzhiyuan purification Engineering Co., Ltd. to learn about the function and maintenance of operation shadowless lamp in laminar flow operating room decoration!
With the continuous development of the medical industry, the operation shadowless lamp is essential and plays the role of lighting. Mechanical installation inspection: check whether there is any mechanical installation fault, whether there is less screw or screw is not tight. Check whether the circlip is stuck, and whether all decorative covers have been covered. Circuit check: first check the system for short circuit or open circuit. If everything is normal, power on test can be carried out. At this time, check whether the external power supply of shadowless lamp is stable. Whether the output voltage of transformer is too high or too low.
Balance arm adjustment: check whether the balance arm matches the lamp head, whether the force on the balance arm needs to be adjusted, whether the angle of the balance arm needs to be adjusted and whether it needs to be adjusted by the adjustment method of the lamp arm. Generally, the screws for adjusting the lamp arm are on both ends of the lamp arm, but there are slightly different from different manufacturers, which can be observed carefully.
Joint flexibility adjustment: it is mainly to adjust the damping screw of the joint. According to the standard, the tightness of the damping adjustment is that the force to push or rotate the joint at any position is close to 20n or 5nm. Adjustment and release of joint limit switch: in order to facilitate the installation and transportation of shadowless lamp, most shadowless lamp products are equipped with lock switch or limit switch in their key parts, such as joints. Now they can be opened.
Check the irradiation distance and depth: most of the shadowless lamps have a irradiation distance of 70-140cm or 60-130cm, that is to say, the shadowless lamps have the best effect in this distance. Check illuminance and color temperature: illuminance and color temperature are actually similar concepts. They are all used to describe the illumination intensity of shadowless lamps. Generally, we think that the most ideal shadowless lamp is that the light is similar to the sunlight, i.e. around 4400-4600k.
The above is the introduction of operation shadowless lamp in laminar flow operating room decoration and maintenance by Shandong jingzhiyuan purification Engineering Co., Ltd. welcome to http://www.sdyhrs.com